A method for accurately counting platelets is described using the Coulter counter model B fitted with a standard 100μ aperture tube. This enables the counter to be used for red and white cell as well as platelet counts using the same aperture tube. The method uses standard equipment except for a small inexpensive electronic speed controller.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.