Aims The presence of cold agglutinin in blood samples can cause a spontaneous agglutination of red blood cells (RBCs) when low temperature occurs. This phenomenon causes a spurious lowering of RBC count on the automated haematological analysers that are detected by incongruous values (≥370 g/L) of the mean cellular haemoglobi concentration (MCHC). A preheating at 37°C can remove the RBC agglutination generally resulting in a reliable count. It has been reported that the same result can be reached by using the optical reticulocyte (RET) channel of Sysmex analysers where the RBC count is not influenced by the presence of cold agglutinin. This study aims to evaluate these data in a larger population, with regard to environmental conditions on Sysmex analysers. We have also evaluated the influence of different thermal pretreatments on the RBC count.
Methods This study was performed on 96 remnants of peripheral blood samples (48 with MCHC in normal range and 48 with MCHC>370 g/L) which have been analysed in different preanalytical conditions on the Sysmex analysers.
Results A preheating of samples at 41°C for 1 min leads to a reversibility of the cold agglutination comparable to the one observed in the RET channel and yields better results compared with 37°C for 2 hours.
Conclusions None of described procedures assure the complete cold agglutination reversibility in every case. Consequently, since the haematological analysers not yet provide reliable parameters to confirm the complete resolution of agglutination, further verification of RBC count accuracy needs to be performed.
- cold agglutination
- mean cellular hemoglobin concentration
- sysmex analyzers
- reticulocytes count
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.