Statistics from Altmetric.com
Many laboratories manage point-of-care services either in their hospital or in their wider locality in association with primary care commissioners. As a result, they tend to be asked to contribute to assessment of the acceptability of new point-of-care systems from the point of view of accuracy and comparability to current systems. This is particularly the case for glucose testing strips, which currently make up 53.7% of the total global point-of-care testing (POCT) market.1
In testing whether systems are acceptable for clinical use, it is usual for ‘pragmatic’ testing regimes to be used. A fully correct ISO 15197:2013 evaluation2 is difficult because it requires collection of capillary blood samples from volunteers with a wide range of blood glucose concentrations, meaning that many healthy individuals and many patients with diabetes mellitus need to be recruited. It is not only difficult to find suitable volunteers but it is also difficult to simultaneously test a wide range of POCT analysers because multiple lancet punctures are necessary as one puncture site rarely stays open long enough for multiple blood drops to be collected. Moreover, there is inherent variability in capillary blood samples collected from fingerprink sites, compounding the problem even further.3 Finally, the use of live volunteers means that regulatory approval for the study would be required from a Research …
Contributors TSP conceived the idea for the editorial, revised and submitted the final version. TMR wrote the first draft.
Competing interests None declared.
Provenance and peer review Commissioned; internally peer reviewed.
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.